GenRad 1693 Precision LCR Meter (RLC Digibridge)
The GenRad 1693 LCR Meter Bridge gives you the best combination of features, in an LCR meter, to meet your most demanding testing requirements. It’s a versatile, flexible instrument with a full range of programmable test frequencies, speeds, and voltages. An easy to understand display show both primary and secondary measurement parameters. The Digibridge is an excellent choice for metrology applications requiring accurate and repeatable measurements. The 1693 is also widely used in production testing, incoming inspection, component design and evaluation, process monitoring or dielectric measurement applications.
The 1693 Precision Impedance Bridge is a sophisticated, patented microprocessor controlled tester that brings new levels of flexibility, simplicity and accuracy to RLC measurement. It’s testing automation at its best with a range of programmable test frequencies and test voltages, as well as automatic limit comparison, automatic parameter selection, remote programmability, automatic binning, and automatic zeroing.
The 1693 Precision Impedance Bridge provides a powerful combination of features designed to maximize productivity in all testing environments.
GenRad 1693 vs. 1689M
The 1693 and 1689M Impedance Bridges are very similar instruments. However, the 1693 offers 2 major advantages over the 1689M.
1. Max Frequency of 200 kHz: The 1693 has a max frequency of 200 kHz, while the 1689M has a max frequency of 100 kHz
2. 11 Impedance Parameters. In addition to the 1689M measurement parameters of R/Q, L/Q, C/R, C/D (series or parallel), The 1693 can also measure R/X (series),G/B (parallel), Z/Angle or Y/Angle
GenRad 1693 Impedance Bridge Additional Uses
GPIB/IEEE Interfaces on the Digibridge Series
IET has an excellent application note on the difference between the new style 1689-9640 GPIB Interface Card and the older style 1658-9620 1689-9620 or 1689-9630 ieee interface cards. This also covers communication with both cards via NI Max. Understanding IEEE/GPIB Interface Cards on the Digibridge Line of RLC Meters
If you are looking for additional information on the 1693 Impedance Bridge, there are a wealth of application notes and tools on this meter that can be found in the Application Notes and Software tab from this page. Any reference made to the 1689 is valid for the 1693
(See datasheet for full specifications)
Measurement Parameters: R/Q, L/Q, C/R, C/D (series or parallel), R/X (series), G/B (parallel), Z/Angle or Y/AngleTest
Frequencies: Over 500 programmable test frequencies (12Hz to 200kHz) 0.01% Accuracy.
Applied Voltage: 5mV to 1.275V (programmable in 5mV steps).
Measurement Speed: ), Up to 19 measurements/second
Accuracy: (Primary parameter) Basic RLC ±0.02%. (Secondary parameter) Basic DQ ±0.0001
Use this link to quickly determine the accuracy for a given measurement on the 1693 Digibridge
1693 Accuracy Calculator
|Parameter||Direct Reading Range||Ratio and DQ in PPM|
|R and [Z]||0.00001Ω to 99999kΩ||0.00010Ω to 9999.9GΩ|
|L||0.00001mH to 99999H||0.00010nH to 9999.9MH|
|C||0.00001pF to 99999μF||0.00010aF to 9999.9F|
|G and [Y]||0.00001μS to 99999S||0.00010pS to 9999.9MS|
|R with C||0.0001Ω to 9999kΩ||not extended|
|X with R||0.0001Ω to 9999kΩ||not extended|
|B with G||0.0001μS to 9999S||not extended|
|D with C||0.0001 to 9999||1 to 9999ppm|
|Q with R or L||0.0001 to 9999||1 to 9999ppm|
|Angle||±0.0001 to 180 degrees||±1 to 999 microdegrees|
GenRad 1693-9700 Impedance Bridge Standard Accessories
1689-0120 1689 Instruction Manual
4200-0300 AC Power Cord
1689-9602 BNC to BNC Extender Cables
NIST Traceable Certificate of Calibration
Address: 5th Floor, 84 Duy Tan Street, Cau Giay District., Hanoi, Vietnam
Tel: +84 (24) 3795 9933
Fax: +84 (24) 3795 9944
Address: No. 54, Street 6, Trung Son Area (KDC Trung Sơn), Binh Hung, Binh Chanh District, HCM
Tel: +84 (28) 5431 6377
Fax: +84 (28) 5431 6378